Rather Like C Plath’s Navistar Professional sextant (see Eighty Years of Carl Plath Sextants, 13 November 2012), the Observator Mark 4 sextant arrived on the scene from Rotterdam just in time to be made obsolescent by the advent of global positioning systems. Like the Navistar Professional, it too had features that were probably a source of annoyance to the practical navigator. Figure 1 shows a general view of the front or left-hand side of the instrument. (All figures may be enlarged by clicking on them. Return to the text by using the back arrow.) There are several striking features: the frame has none of the traditional bracing; there are no shades to be seen; the index arm is on the back of the frame;and the very squat telescope is almost as one with the frame, as well as having a variety of knobs.
On turning over the instrument to see a rear view (Figure 2) it becomes immediately apparent that it is a little awkward to do so with one hand because of the shape of the frame. The handle is then seen to be a very stout alloy casting attached only at the top by three Allen screws, and the micrometer mechanism is seen to be totally enclosed within a U-shaped casting attached by three screws to the lower end of the index arm. A further knob is revealed on the back of the telescope.
Figure 3 shows the structure of the frame, a heavy, monolithic aluminium alloy casting 15 mm thick, with the index arm bearing and the rack machined directly into it and with the engraved arc screwed and glued to the limb. The whole instrument weighs 1.9 kg (4 lbs 3 oz), rivaling the heaviest of traditional bronze sextants.
The word “bearing” is often taken to mean the rotating part plus its enclosure, but strictly, the central rotating part is the journal and the enclosure in which it rotates is the bearing; and I have kept to this traditional nomenclature in Figure 4. The large diameter journal carries the index mirror bracket which is attached to it by two screws from below. The large diameter bronze journal rotates in the bearing which, as in the Soviet SNO-T sextant and the Freiberger sextants, is machined directly into the frame and which seems to be anodised for hardness. The upper end of the index arm and a retaining washer are located by a central spigot on the journal and attached by means of two countersunk screws from the rear. The index arm is a piece of 25 x 5 mm rectangular aluminium alloy stock.
The index mirror measures 25 x 50 mm and has the usual springs to retain it in its bracket and an Allen screw on the rear to adjust for perpendicularity. The horizon mirror (Figure 5) is of the same size and is fully silvered. That is to say, there is no plain glass portion through which the horizon is viewed, but rather it is viewed directly. With a half-silvered mirror, about 10 percent of the light from the index mirror, assuming it is of a size to do so, is reflected off the front and back surfaces of the unsilvered portion of the glass and the effect when the telescope is a Galilean type is to widen the area of view where the direct and reflected images coincide. This makes it rather easier to judge when the observed body and the horizon coincide. The top of the bracket, otherwise identical to the index mirror bracket, has been machined away so that the view over the edge of the mirror is unimpeded. There are the usual springs and screws to adjust for side and index error. Both brackets are very robust and strongly mounted, and it would take considerable force to displace them.
The telescope, which contains the shades within it, is perhaps the most unusual feature of the Observator Mark IV sextant. Figure 6 is a cross section through the telescope, modified from Figure 3 of the patent document for the sextant, filed in December 1982. The telescope is a Galilean type in which, in effect, light from the left half of the field stays in the left half of the telescope while light from the right half of the field stays on the right. This is perhaps easier to understand by covering half of the objective lens of such a telescope, when it will be seen that the corresponding half of the field will go dark, but if this is repeated with a Keplerian or inverting telescope, the field will simply go dimmer, while the full field of view is retained.
Observator took advantage of this characteristic of the Galilean telescope by inserting a filter or shade in each half of the light path inside the telescope, so that the intensity of light from the observed body and from the horizon could be independently controlled at the telescope by means of curved filters of continuously varying density, from light orange to deep red. To make quite sure that no light from the body could pass through the lighter horizon shades a central flat diaphragm lies along the axis of the telescope, as seen in Figure 7, which also shows the objective lens in its cell.
Figure 8 shows the form of one of the shades, in which a piece of coloured photographic film is held in a curved holder which can be rotated by means of a large external knob to bring different parts of the film into the light path. There is also a semi-circular notch in the bracket that allows light to pass unimpeded.
Focusing is carried out by moving the objective lens axially by means of an external knob and a crank that engages with a slot in the objective lens mounting (Figure 9). Magnification is about four times and colour correction is not complete, with coloured fringes being visible on a bright horizon, though as soon as the filters are deployed this is unlikely to be a problem. Operation of the focus and shades controls is easy.
The main advantages claimed for the telescope are that the shades are kept free from dirt and damage, inside the telescope. These advantages have been bought at the cost of complication, balanced by the negligible cost of the photographic film, but the film is practically in the focal plane of the objective lens, and in my example the index filter had been damaged by heat. This was easily remedied by reversing the film edge for edge in the bracket.
Figure 10 shows the micrometer mechanism detached from the index arm. End play of the worm in its bearing is taken up by a spring pressing on a ball bearing. The bearing of the worm is screwed to a bronze swing arm that swings about a pivot under the influence of the release catch, the end of which is just visible in the Figure 10 above the worm. The catch passes through a slot in the base of the micrometer mechanism casting and acts as a crank to rotate the swing arm and its attached worm out of engagement with the rack. Many sextants when dropped suffer damage to the release catch and shades, but the one is very robust and the other out of danger inside the telescope. Although in a classical sextant there may also be a worry that the frame has been bent, this is unlikely in an alloy-framed instrument and it is difficult to imagine that it might happen to the frame of this instrument.
The exploded view of the mechanism in Figure 11 shows how a leaf spring beneath the swing arm presses the worm into engagement with rack. The 15 mm diameter drum is divided into minutes and, unlike that of the Plath Navistar Professional, is easy to read, though there is no provision made to illuminate it.
The case is made with solid mahogany walls with corner rebate joints and plywood top and bottom. The hinges and other hardware are of plated steel. It is provided with an oil bottle and an Allen key for adjusting the mirrors. The index arm must be set at 45 degrees in order for the sextant to fit face down in the case, a minor annoyance only. If it is necessary to set it down on a flat surface, it tends to rest on the telescope and micrometer, both of which are rugged enough to stand it, rather than on a mirror or shade as in conventional instruments.
This is a rugged, practical instrument with few weak points other than the fully silvered horizon mirror and poor colour correction of the telescope.
If you have enjoyed reading this account, you will probably enjoy reading my book on the structure of sextants, “The Nautical Sextant,” and if you are interested in the instruments of navigation, my other book, “The Mariner’s Chronometer,” may also be of interest to you